Insole & Insock Water Absorption & Desorption TesterAmadeTechAdmin2021-09-17T08:07:06+00:00
Footwear Water Absorption and Desorption Tester China Manufacturer
Insole & Insock Water Absorption & Desorption Tester is intended to determine water absorption and desorption of footwear’s insole and insock.
A test specimen is positioned on a wet base plate and is subjected to repeated flexes under a given pressure(in the same manner as the insole of a shoe during walking). the water absorption at the end of test and the water desorption following the tests are determined.
ISO 22649 Footwear-test methods for insoles and insocks-water absorption and desorption
EN ISO 20344 section 7.2 Personal protective equipment-test methods for footwear determination of water absorption and desorption of insole and insock
EN ISO 20347 section 5.7.3 Personal protective equipment-occupational footwear
ISO 5404 Leather-physical test methods-determination of water resistance of heavy leathers
EN 12746 Footwear – Test methods for insoles and insocks – Water absorption and desorption
Main Technical Parameters
Diameter (120 + 1) mm, Width (50 + 1) mm
20 ± 1 cycles / min
50 ± 2mm
80 ± 5 N
Water Flow Rate
7.5 ± 2.5 ml / min
(110 ± 11) × (40 ± 1) mm
0 ~ 999 h / 59 m/ 59 s
1 phase AC 220 V 50 / 60 Hz or specified
Procedure of Water Absorption Test
Weight the test piece to an accuracy of 0.001 g.
Position the cotton gauze on the platform.
Mount the test specimen into the apparatus, ensuring that the surface in contact with the foot contacts the platform covered with the cotton gauze. Attach the narrow end to the platform and the roller, and a force of (80 ± 5) N is applied.
Open the valve to enable water flow and adjust this to (7.5 ± 2.5) ml/min over the platform.
Actuate the machine and note the time. Run the test for 1 hour and stop the water supply 1 min before stopping the machine.
Remove the specimen and weigh it to the nearest 0.001 g, recording its mass.
Recondition the test piece by leaving it on a flat waterproof surface in a controlled environment for a period of 24h, then reweigh the specimen to the nearest 0.001 g.
The above is just for reference. Please strictly follow the procedure of the applied standard to carry out the test when you get this testing machine.
Send an Inquiry
Please feel free to contact us for more details on the product, price, lead time, payment terms, shipment methods, etc. Amadetech sales specialists will respond within one working day.